1. Heating plate: be made of 304 stainless steel, test area 900*450mm, surrounded by protective layer, width 80mm
2. Heating tube: be made of SUS304 stainless steel, the length of 2M, nominal diameter of 88mm, 108mm, 133mm three types
3. Temperature measurement system: the diameter of the thermocouple wire is 5.0mm, the range of measurement temperature: 0-1250 ℃; using PLC + module acquisition temperature.
4. Control part: PLC module, computer control real-time display hot surface temperature curve, output test report.
5. Heating rate: 5 ℃ / min or 3 ℃ / min can be set.
6. Temperature acquisition: collected once every 2min, together with curve shows.
7. Thermocouple response time: <0.5s.
8. Temperature error: ± 0.5 ℃.
9. Timing error: 1s / h.
10. Computer: 1 set (at least meet with these following requirements: CPU: INTEL I5-III on behalf of the above; Memory: 4G or more; Hard disk: 500G or more; with the serial port: control equipment operation; Display: 20 inches or more LED Liquid crystal display, 1600 * 900 resolution above; windows xp system.), A printer. Test software set.
11. Equipped with flat feet and feeler: measure warpage.
12. Equipped with two sample cans.
- Quality Engagement
- Easy change and return
- Delivery Avaliable
- Favorable payment